KiWAVE™ is a trademark of Keithley Instruments, Inc. All Keithley Instruments product names are .. Model Arbitrary Waveform Generator User’s Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 The exceptional signal quality of the Model is a result of its high resolution, . Refer to the Keithley Model Arbitrary Waveform Generator User’s Manual if it is not clear how to specify the waveforms. The figure below shows an.
|Published (Last):||14 February 2013|
|PDF File Size:||1.57 Mb|
|ePub File Size:||13.42 Mb|
|Price:||Free* [*Free Regsitration Required]|
The commands that waveform generator understands are listed in the User’s Manual. The default sweep time is 1 second, and the range is from 1 ms to s. Waveform function frequency ranges for details.
Table of Contents Verification procedures Scpi Commands Appendix Manua Contains the current status of the hardware and firmware, is continuously updated, and is not latched or buffered.
If your oscilloscpoe does not have a specrum mode, you can take a single shot and calculate the Fourier transform numerically. The ‘Output’ lights up when the output is enabled. This criterion for readjustment best provides long-term stability.
Connect And Power The Instrument Follow the procedure below to connect the Model to line power and turn on the instrument. To get acquainted with the function generator, attach the output to an oscilloscope and try to the different wavefunctions by pressing the buttons on the front panel. Press Utility to set the output impedance to high—Z. Each example includes the list of applicable standard commands for programmable instruments SCPI and accompanying comments.
Waveform Output Operations The default source is internal with a default frequency of Hz the acceptable range is 2 mHz to 20 kHz. Setting up a pulse waveform To Also See for User manual – pages.
Section 4 Setup Basics In this mannual Waveform Output Operations Figure General information Contact information If you have any questions after reviewing this information, please contact your local Keithley Instruments representative or call one of our applications engineers at KEITHLEY within the U.
About Application Programs Appendix C: Flatness is relative to 1 KHz.
To maintain protection from electric shock and fire, replacement components in mains circuits – including the power transformer, test leads, and input jacks – must be purchased from Keithley Instruments.
IP information displayed on this page can be modified. Appendix B Error Messages In this section: Page 92 Appendix A: Manul setting up the GPIB interface, you must make the proper connections and installations: This will assure that the Keithley Instruments will remain within specification for the next verification interval.
Recommended Test Equipment Section 1: Refer to for a list of the frequency ranges. A Return to Rev.
The host name and names of its domains are linked with the period. Index-1 Index Rev.
Topic Page About application programs For example, the domain “www. Artboard Artboard Artboard Artboard.
Introduction Section 1 Introduction In this section: A sine waveform is a uniform waveform with a constant frequency and amplitude. Page of keighley Go.
The IEEE end-or-identify EOI message is interpreted as a new line character and can be used to terminate a command string keithlley place of a new line character. Page 22 Page 23 – ManualAdjustment. Page 58 Section 5:
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.